If You Want to Know and Purchases Electronic Extensometers. Then, You Can Contact us Shortly and Check Features. We added here detailed specifications for the Electronic Extensometers (Model : FEE-5).

- This instrument is to be used on Tensile or Universal testing machines to find out Proof stress & Young’s modulus values. In case of many brittle materials such as high carbon steels, alloy steels, light aluminium & magnesium alloys, it is difficult to get yield values. For such materials stress corresponding to a certain allowable amount of plastic deformation is termed as proof stress say 0.1% or 0.2% proof stress.
- The measuring range is upto 5mm & resolution is 0.001mm.
- Standard two fixed gauge lengths of 25 & 50 mm is provided. Different sizes gauge lengths can also be given at an additional cost.
- If this extensometer is ordered with a computerized UTM, then a special menu driven application software is provided,
- which accepts load & extension values, & evaluates proof stress values.
- The measurement accuracy satisfies all requirements of class-I as per IS:12872-1990.

**Technical Specification – For Electronic Extensometer – Model – FEE – 5**

Accuracy | Class 1 as per IS 12872 / ISO 9513 |

Type | Strain Gauge type |

Out-put | 0.3 mV / V / mm. (approx) |

Excitation | 10 V. DC Nominal. |

Measuring range | 5 mm. |

Nominal resistance of bridge- | 350 ohm |

Gauge length | 25 mm/50mm by means of extender |

Resolution | 0.001 mm |

#### Related posts:

Bottom Discharge Type Diverter

Side Discharge Type Diverter

Portable Dynamic Hardness Testing Machines (Model : DHT-6)

Dense Phase Pressure Conveying System

Rockwell Cum Brinell Hardness Testers (Model : MRB, MRB-250)

Transverse Testing Machines for Rails (Model - FCT- 2000, FCT- C-2000)

Vertical Type DSP Based Dynamic Balancing Machines (Model : FVBM)